Executive internet pharmacy in Sydney where you can buy propecia australia online. Para compra levitra puede ser visto como un desafío. Aumenta Smomenta, y todos los que se poco a poco abrumado, como es lógico, cada vez más hombres están diagnosticados con disfunción eréctil.
A D M I E
MEASUREMENTS AND INSTRUMENTS SECTION
PROGRAMMABLE DISPLAY & DATA RECORDERS
This technical description covers the technical characteristics and the required tests
for Programmable Display and Data Recorders to be installed in Transmission
These Programmable Display and Data Recorders will accept at its inputs analogue
DC signals for measuring, recording and displaying on TFT-display simultaneously at
least six (6) values, by selection, I, V, W, VAR, HZ and cosφ. All these data must be
stored to a detachable circular buffer for duration of at least 90 days.
The technical characteristics and tests of these Programmable Display and Data Recorders must be in accordance with DIN 43700, EN 61010, EN 60529, EN 50081-1, EN 5082-2, EN 61326.
Number of Inputs:
± 20 mA DC, 4-20 mA DC & ±50 V DC for each
Inputs Accuracy Voltage accuracy:
± 0.2% up to 10 V DC & ± 0.1% greater than 10 V DC Current accuracy:
± 0.2% with external shunt resistor
Four, by selection, programmable in steps up to 0.5 mA
Current circuit < 10 Ω ± 5%
Digital indications and recording:
Colored of at least 12 colors.
Colored touch screen of high definition LCD (TFT) and of industrial
resistance. All settings, operations of the data recorder will take place via the touch screen.
At least 5.5” (diagonal)
at least 320 X 240 pixels (QVGA)
Screen update rate:
Every 1 sec
Internal Storage Media:
internal storage media at least 400 MB, for 12 inputs
and sample rate 1 sec, data storage for at least 90 days
External Storage Media:
PCMCIA at least 512 MB, possibility of extracting
Serial Port USB or RS232 or RS485
Serial Port RS485 Modbus RTU and
It will be of calendar function, adjusted manually as well as with a
program via a portable PC. Power supply via a lithium battery with life duration of at least 10 years. It will also have the possibility of automatic change of summer hour.
1. Numerical indicators in full screen displaying
2. Vertical or Horizontal charts. Chart speed for 1 mm/h up to 120 mm/h 3. Vertical Bargraphs 4. Horizontal Bargraphs 5. Maximum and minimum values.
Values, date and time, channel and
Adequate number of event markers for limit monitoring,
Appearance and recording of values via mathematic
90 ÷ 260 V AC/DC 50 VA max
Programming security codes in several levels.
Flush-mounting back connected panel for panels of thickness
2-20 mm according to DIN 43700. The back case shall be made of galvanized steel or proper synthetic matrerial.
-10° C ÷ 50° C
Accuracy and stability tests of the measured values.
Electromagnetic compatibility tests for industrial environment according to EN 61326 Class B.
According to EN 61010-1 category II, pollution degree 2. Test voltage between analogue input channels and earth 300 V DC or peak AC and 100 V DC or peak AC among the input channels.
Protection class test of front part IP 55 and rear part IP 20.
Dielectric test of inputs and certificate print.
A. ADDITIONAL PURCHASE REQUIREMENTS
Detailed manuals for the operation and programming in English. 2.
Along with delivery of the Programmable Display & Data Recorders it should be
also offered necessarily, according to the specified quantity by the inquiry,
PCMCIA cards. 3.
Suitable software in Windows XP environment for programming the display and
data recorders, reading and processing the data of multiple use licenses. The
software will have the possibility of data conversion to excel sheets. 4.
Suitable software in Windows XP environment for telemetering data via PSTN
modem of multiple use licenses.
B. ADDITIONAL OFFER REQUIREMENTS
Along with the technical offer, it must be submitted CE and ISO 9001 conformity
A sample from type tests certificates by an internationally recognized private
or state laboratory, has to be submitted with the technical offer. 3.
All bidders must in detail refer to every single paragraph of the Technical
Description, otherwise it will constitute a sufficient reason for the rejection of the offer.
Jesse Hutt, Amy Hewitt, Katherine (Jeffrey), Danielle Sullivan (fresh), Flow Wolf (fresh), Cate Smock, Jessi Kileene (frosh), Diane Gray (10th), Cari Harrison, Kristina Monroe,Meghan McGrath, Tiare Flora, Carin Garcia, Kate McGrath, Jane Katz, Lynn Arava, School Counselor – Report on Mental Health Survey Jake noted that we are in 5th year at AHS, and this is first year we have had a c
“ RELACION ENTRE EL TRATAMIENTO ANTIGIARDIÁSICO Y ESTADO NUTRICIONAL EN PREESCOLARES DEL ASENT AMIENTO HUMANO ALTO TRUJILLO DEL DISTRITO EL PORVENIR” Autores: Bazán Cayetano Johana, García Saucedo Sheyla, Campos Florián Julio Cátedra de Farmacología, Escuela de Farmacia y Bioquímica, Facultad de Farmacia y Bioquímica, Universidad Nacional de Trujillo, Perú. El present